YY/T 1198-2013: Translated English of Chinese Standard. (YYT 1198-2013, YY/T1198-2013, YYT1198-2013): Aspartate aminotransferase diagnostic kit (IFCC method) [Tips: BUY here & GET online-reading at GOOGLE. Then, if you need unprotected-PDF for offline-reading, WRITE to Wayne: Sales@ChineseStandard.net]
This standard specifies the determination principle, requirements, test method, marks, signs, instructions, packaging, transport and storage of aspartate aminotransferase diagnostic kit (lFCC method). This standard applies to the quality control of aspartate aminotransferase diagnostic kit (lFCC method) This product is used for in-vitro quantitative determination of the aspartate aminotransferase activity in human serum or plasma.
What people are saying - Write a review
We haven't found any reviews in the usual places.
Mark Label and Instructions
Packaging Transport and Storage
absorbance A2 Absorbance of reagent according to Formula aspartate aminotransferase diagnostic Between-batch precision Calculate the mean Calculate the relative calculated through measuring certified reference material coefficient of variation comparison method concentration of human correlation coefficient coupled reaction determine the same-batch diagnostic kit lFCC diluted concentration xi Expiry date expiry of validity following contents GB/T high and low-value human serum sample Installed capacity lFCC method limit of linear linear interval linear regression low-value quality control Marking for Handling mean value measured concentration measured results yi measured value meet the requirements NAD+ NADH has characteristic Oxaloacetic acid Product name Production batch number quality control material Rate of change rate of NADH reagent blank absorbance reagents kits recovery rate relative deviation Repeat the determination Repeat the test results shall meet serum whose value single kit packaging standard deviation standard solution test results Test the reagent volume of standard wavelength of 340 Within-batch precision